Mouth alcohol is another potential source of error with the Intoxilyzer 5000 used by the Bryan, Texas Police Departments. If alcohol is regurgitated into the mouth (by burping or belching) and absorbed into the mouth tissues, it will effect the breath test result by overestimating the amount of alcohol attributed to the breath sample. Even the act of forced exhalation required to provide a breath sample can cause the rib cage to squeeze and force stomach gases into the breath resulting in a false overestimation of breath alcohol content.
Generally, it takes about 10-12 minutes for mouth alcohol to dissipate and not effect the test. However, if a person has belched within about 10 minutes of the test, the mouth alcohol will falsely contribute between .01 and .03 to the result. The breath test operator protocols require the officer to “be in the presence of” (but not observe) the DWI suspect for 15 minutes to assure they don’t belch or place foreign objects into their mouth effecting the test. As a practical matter, though, the officer is rarely paying much attention to whether the suspect is burping or belching. They are typically busy filling out paperwork, driving the suspect to the police station, talking with other officers, or getting the Intoxilyzer warmed-up for the test. If a person burps silently there’s no way the officer is going to detect it.
The Intoxilyzer 5000 does attempt to detect mouth alcohol by what is called the “slope detector.” The slope detector is a computer program designed to measure the rate at which alcohol concentration changes inside the Intoxilyzer 5000. If the concentration changes too fast (i.e., too “steep” of a slope), the machine registers an alert and the test is aborted. The slope detector must be “satisfied” before the Intoxilyzer 5000 will produce a result. The problem is that empirical studies show the slope detector does not always work as advertised. Moreover, Texas does not have a standard procedure approved for use by the technical supervisors to verify the slope detector feature of the Intoxilyzer 5000 is working properly and conforms to the manufacturer’s certification.
At trial the technical supervisor will generally testify: (1) they don’t know what process was followed in developing the slope detector program; (2) they don’t know of any treatises or literature that support the use of the process followed to develop the program; (3) they don’t know whether the program has been published or offered for publication in any peer-reviewed journal; (4) they don’t know whether the program been described in any peer-reviewed journal; (5) they don’t know whether any other scientists use or recommend use of the program; (6) they don’t know whether there’s a known or potential rate of error; and (7) they don’t know whether there are any studies showing the slope detector actually works as advertised.
You need a Bryan DWI Defense Lawyer who understands the mouth alcohol problem and can use it to your advantage when defending against a Bryan DWI. Call Now.